Accurate entry of "M" data is required for accurate interpretation of the cycle.
You have likely entered data incorrectly or already measured a temperature. Delete data and reset the device.
To delete data: Proceeding from the Main Menu, select the Tool icon. On the Tools menu, select the Trash Bin icon. Press the centre of the toggle button repeatedly until the red line moves completely to the left side of the display and the Lady-Comp beeps indicating the reset is complete. The sample temperature data will be restored to the device.
To confirm deletion of data: Select the Thermometer icon from the centre of the main menu. Fahrenheit display will show sample data: “97.97 alternating with 14 (day of the cycle)”.
Proceed to re-enter “M” history, BMI and reset and re-activate the alarm.
Proceeding from the main Menu:
1. Select the M icon; You will see today`s date in the bottom right side of the display .
2. Press the Left Arrow to reverse the date to the first day of M, the furthest date back.
3. Press the Up Arrow to Check the box.
4. Press the centre of the toggle button to confirm your entry.
5. Press the Right Arrow to proceed to the next more recent first day of your menstrual history.
6. Press the Up Arrow to Check the box, the centre of the toggle button to confirm and proceed to next date by pressing the Right Arrow. Continue until all historical dates have been entered up to the most recent date.
Proceeding from the main menu, select the Thermometer icon. The accurate Cycle Day is displayed beside the date on the lower right side of the display. The number (Cycle Day)must be accurate to the number of days since the first day of your last menstruation.
If you have been measuring the temperature each day you can only remove or add Menstruation data within 3 days of the error.
If you have not measured any temperatures since the error, you may be able to correct more than 3 days back. Proceeding from the main Menu, select the M icon. To remove the current day`s M data, press the Down Arrow to uncheck the box, then press the centre of the toggle button to confirm. Now press the Left Arrow to display yesterday`s M data, if the M box is checked, proceed as above and continue to remove all incorrect M data. Now move the date by pressing the arrows to the accurate first day and press the Up Arrow to check the box, press the centre of the toggle button to confirm. Now select the Thermometer icon and make sure the accurate Cycle Day is now displayed beside the date on the lower right side of the display.
If you are unable to correct “M” data because more than 3 days have passed, you must exercise caution and avoid unprotected sex for the remainder of the cycle until you confirm the next accurate menstruation date and correct the Cycle Day. Use an alternate form of contraception (non-hormonal, such as barrier methods) or abstinence on both red, yellow and green light days for the remainder of the cycle.
Enter only one day of “M” when confirming historical data.
If you are entering current menstruation data while measuring the daily temperature, you must confirm “M” for at least 3 consecutive days. You can confirm “M” for the length of your period for your own reference, but you must confirm at least 3 days for confirmation of actual menstruation. If you are a new user, you may decide to Delete all stored data and reset the device as described above. The daily fertility diagnosis will not be reliable if the Cycle Day is not accurately indicated on the Daily Temperature display.
The thermometer symbol must be illuminated on the Time and Date display when the alarm sounds in order to measure the daily temperature reading. The thermometer symbol will illuminate 3 hours before the previous day`s measuring time and remain illuminated until three hours after or until a daily reading is measured.
The temperature reading time frame (6 hour) will need to be set by activating the alarm for the initial reading and re-set by activating the alarm if you miss three consecutive temperature readings(intentionally or not). It must be re-set before Midnight on the missed day so that you are able to measure the temperature again the next day.
To re-set the Temperature Reading Time Frame you must re-activate the alarm bell:
Before Midnight, the day of the missed reading; Proceeding from the main menu:
Select the Alarm Bell icon. Select the Alarm Bell, again. Select the Down Arrow to uncheck the box. Press the centre of the toggle button to confirm the entry. Exit this display by selecting the Back icon. Select the Alarm Bell again. Select the Up Arrow the check the box. Press the centre of the toggle button to confirm and select the Back icon and return to Main menu.
There are a few factors that can interfere with the temperature measurement.
First make sure the sensor is working normally. Open the device and make sure the sensor plug is fully inserted to the socket by pressing firmly on the plug. Now try a test measurement. Select the ‘wrench’ icon from the main menu. Place the sensor tip under your tongue and select the ‘thermometer/C’ icon on the tools menu. Do not press the button again once the thermometer symbol is flashing or it will cancel the test. Wait for 1-2 minutes to see if the measurement is confirmed with a beep.
Another reason for failed temperature measurement is related to the Measuring Time Frame. If you have missed more than 3 days of measurements, the Measuring Time Frame may have been deleted and you need to reset it by un-checking the Alarm Bell box, exiting the menu with the arrow back button. Then select ‘alarm bell’ icon again and re-check the box. This will reactivate your 6 hour Measuring Time Frame.
A third reason for failed measurements may occur if you are trying to measure outside of the Measuring Time Frame. Only one daily measurement is permitted and it must be done within three hours of the previous day's measuring time. If you measured at 6 AM yesterday, today you would only be able to measure a reading between 3 AM and 9 AM. You can confirm you are within the Measuring Time Frame if the Thermometer symbol is illuminated on the home screen.
The failed reading could also be related to how the measurement is activated in the morning. When you wake-up in the morning, you should be able to see the thermometer symbol illuminated on the display to confirm you are now within your Measuring Time Frame. When the alarm rings, you should only press the center button once to stop the alarm. Then place the tip of the sensor under your tongue and press only once to activate the measurement.
Once the thermometer symbol begins to flash, do not press the button again or the measurement will be cancelled.
If you wake-up before the alarm and the thermometer is illuminated, measure before getting out of bed. In this case, place the sensor under the tongue and only press the center of the toggle button once to activate. Wait for confirmation of measurement with a beep.
If a reading is secured each day, there is no need to re-activate the bell. It will continue to record the temperature each day without any further action.
Although you can turn off the alarm after the first reading, it is not recommended for women who are “new” to the method. The Alarm Bell is a reminder to measure and missing readings results in increased red and yellow light days. If you forget to measure on three consecutive days, you must re-activate the alarm bell before Midnight, to reset the Measuring Time Frame for the next day’s reading.
It is recommended to allow the alarm to wake you until the routine becomes established and it is less likely you will forget and miss a daily reading.
In a typical 28 day cycle, a woman can expect to receive between 8-10 red light days per cycle and the remainder will be green. Most of the red lights will occur in the days leading up to ovulation and for 48-72 hours after it has been confirmed.
An increase in red lights results when the ovulation is not detected in the cycle. Three factors can contribute this:
1. Missed and Inaccurate Basal Body Temperature Readings: Temperatures must be measured immediately upon awakening from at least 3 uninterrupted hours of sleep. You should not get out of bed, drink fluids or engage in any activity within 3 hours of the measurement. If you miss a daily reading, especially near to ovulation time, the Lady-Comp cannot confirm the timing of fertile days in the cycle. You may not be aware that you missed the readings. To confirm that a temperature has been measured each day, proceed from the main Menu:
Select the Thermometer icon:
Press the Left Arrow repeatedly to display previous day's temperature data.
2. Anovulatory Cycles: It is common for women to experience an occasional cycle where no ovulation occurs. When this occurs, the Lady-Comp cannot isolate the fertile days of the cycle and will issue red and yellow lights as a precaution.
3. Ovulation not detected through typical temperature patterns: Some women do not experience a dramatic shift in the basal body temperature readings with ovulation. If this is the case, most women can still use the Lady-Comp effectively because it does measure the BBT very accurately to the 100th degree. In order to detect ovulation, it would be recommended for her to secure an accurate BBT each day at the same hour of day. Also, avoid any missed daily readings.